How do you use Bair Hugger?
- Application Steps. 3M™ Bair Hugger™ Multi-position upper.
- Place. Once surgical positioning is complete, place the blanket on the patient.
- Bend. Bend and conform the blanket on or around the patient to cover the greatest amount of skin surface area.
- Remove. Remove adhesive strip backing.
At what temperature do you use a Bair Hugger?
36.0° to 37.5°C2
Bair Hugger Temperature Management Solutions provide the widest range of best-in-class patient warming products to help clinicians keep a patient’s core body temperature within the normothermic temperature zone of 36.0° to 37.5°C2,3 and safeguard patients while improving outcomes.
What side goes down on bear hugger?
Always place the perforated side (the side with small holes) toward the patient.
How does the Bair Hugger work?
The Bair Hugger warming unit is attached to the warming blanket or warming gown by means of a flexible hose. Warm air is generated in the warming unit and flows through the warming unit hose and into the blanket or gown.
What is the model number of the Bair Hugger™?
Ohřívací jednotka 3M™ Bair Hugger™, model 775 – Návod k obsluze Reset při zapnutí napájení ohřívací jednotky
How to clean 3M Bair Hugger model 775 warming unit?
3M™Bair Hugger™ Model 775 Warming Unit – Operators Manual General Maintenance and Storage Cleaning Instructions 1. Disconnect the warming unit from the power source before cleaning. 2. Cleaning should be performed in accordance with hospital practices for cleaning OR equipment.
What does the Bair Hugger service manual 0470 warning mean?
BAIR HUGGER SERVICE MANUAL Model 500/OR Warming Unit BAIR HUGGER* SERVICE MANUAL 0470 Warning: Electrical Shock Hazard. There are electrically live parts within the warming unit when it is connected to a power source, even when the switches are in the OFFor STANDBY position. Please forward to Biomedical Engineering Department